|Cigarette sample- Measurement Report|
The sample was successfully scanned using a ConoLne-100 in order to estimate the radius of the cigarette.
|Measurement range (mm)||30|
|Line length @ center (mm)||15.5|
Lateral resolution [X*Y]: 100μm X 100μm𝜇𝑚
Scanning axis: Y axis
Measurement frequency: 100 lines/sec
Mirror type : T
Laser power: Max
Auto exposure: on
Scanner system: Application system
Several radius along the cigarette:
Several cross sections of the cigarette were analysis.
1th Radius and deviation:
As can be seen the Radius is 3.702mm with radial error (STD) of 95um.
2th Radius and deviation: As can be seen the Radius is 3.775mm with radial error (STD) of 63um
3th Radius and deviation: As can be seen the Radius is 3.679mm with radial error (STD) of 70um.
4th Radius and deviation: As can be seen the Radius is 3.691mm with radial error (STD) of 65um.
ResultsAs can you see the radius of the cigarette can easily be measured using ConoLine100 using T mirror only.
The cigarette I measured is not a new one but It can easilly be obtain that Optimet sensor can measure it without any problem.
The Cigarette radius was changes between :3.679 to 3.775 , it also can be seen that the shape was not as round as it is when is get out of the manufatoring system.
Download Cigarette Scanning Report file (pdf)